BK 879B METER REVIEW REPAIR PROBLEMS Err02 error02
I dont recomend anyone buying BK meters. Their service dept is very
poor. They dont have schematics available for their products. They
offered to replace the guts of my meter for $170 thats crazy. HEY BK,
you sell meters so go ahead and assume most of your customers are techs
and can repair the meters themselves. sheesh.
They claim error 02 is from measuring uncharged caps but I turned my meter on after sitting all night and it died.
OK thanks to some bosneian techs I got this working
U603 PFKI 595-TPS60400DBVR
D201,D301 komponenta SMAJ5.0CA
D304,D305 komponenta 1N4007 smd
HERES THEIR LINK http://www.elitesecurity.org/t479669-3